T2514 Ethernet sensor is designed for measurement of barometric pressure. Display is possible to switch off. Online transmitter is designed for use in on-aggressive environment.
Modes of communication Modbus:Modbus TCP protocl enables to read measured values, set alarm limits, adjust the probe, read firmware. Telnet: Port 9999 enables to set alarm limits (lower and upper limits for T, RH, Tdp, hysteresis and time delay), e-mail addresses, SNMP addresses, probe description, refresh of www pages (10s to 65535s), select type of www pages, set storing interval to history (10s to 65535s), enable each communication channel. Capacity of the history memory is 100 sets of measured temperature, humidity, pressure+computed value. Password protection of this port is enable. Automatic IP address assignment from DHCP server is also enabled. www pages: User selectable design of www pages enabling to dispaly curves of measurement history.User can design the look of www pages and select values to display. SNMP: It is possible to read actual values and alarm limits. In case of alarm creation warning message (trap) is sent to IP addresses defined by the user (maximum three addresses). E-mail: In case of alarm creation warning e-mail message in sent to addresses defined by the user (maximum three addresses). Syslog: Transmmitter enable to send text messages to selected syslog server after different events appear.E.g. after transmitter restart, alarm activation, communication error with SNTP, write to transmitter via mdb,sntp, afterfirmware change, after alarm termination, after communication error with SOAP server.
Operating temperature range: from -30 to 80°C (is recommended to switch OFF the LCD over 70°C).
LAN Connection: RJ-45
Dimensions: 89 x 73 x 37 mm
: 9-30Vdc, max 1W
Power connector: axial, diameter 5.5 x 2.1 mm